Last updated: 31 July 2007
This page is for user comments and information specific to the Meade AutoStar Suite. If you have any comments, suggestions, questions or answers to questions posed here, e-mail them to me and I'll post them. Please use an appropriate Subject Line on your message. Thanks. (The Lunar Planetary Imager, LPI, and Deep Sky Imager, DSI, are covered in the Helpful Information - Astrophotography area.)
Subject: PEC training Sent: Tuesday, July 31, 2007 11:44:14 From: Stephen Olson (firstname.lastname@example.org) Just got my scope back from Dr. Clay. What a great guy and great service. He suggested that since I added my AT66 and telrad to my scope that I have to do PEC training. I never did this, and just don't what to mess things up. Here is what meade says to do in their manual: 1. Select "PEC Train" from the Smart Drive menu (in the "Setup: > Telescope" menu) > > and press ENTER. > > 2. Look through the reticle. Use the Arrow keys to keep the star > centered in the reticle > > cross-hairs. Autostar displays a count of the 150 positions of the > worm gear. When > > the count reaches 150, one cycle is complete. A full cycle takes about > 12 minutes. > > 3. Press MODE to exit. Is there anything that I am missing? In step 2 it asks to use the arrow keys to keep the star centered. Do I have to look into the eyepiece for the next 15 minutes to do this? Also if I use the DSI camera to do this does the mount have to be connected to the computer for guiding or can I just use it unguided???? Thanks for your help one more time! I owe you one! Steve www.theurbansky.com
From: richard seymour (email@example.com) That's pretty much the procedure... yes, you need to continuously guide the scope during the procedure, but you can have the camera do it for you. If you're expecting the camera to handle the guiding/centering during the training, it does have to be talking Camera->(usb)->PC->(serial)->Autostar. You can improve the PEC performance by doing PEC Update cycles after the Training. They're just like Training, except that the new data is averaged in to the old data to smooth out the "noise". Some folks find that the tracking rate slows down after doing a PEC Train (and then having PEC engaged). Setup>Telescope>TrackingRate>Custom For the PEC data to remain in sync with the worm angle, you -must- PARK the scope at the end of every session. As long as you do that, you won't have to retrain. You can unclamp the clamps and remove the OTA without losing the PEC sync, as long as you Utilites/PARK. If you'd like to -see- your PEC data, you can use Andrew Johansen's PEC Editor from http://members.optusnet.com.au/johansea have fun --dick
Subject: Autostar #494 Sent: Friday, July 27, 2007 06:10:42 From: jon + jen venditti (firstname.lastname@example.org) Mike, first let me say I really find the Mighty ETX site very useful. I am a newbie but I am learning fast. I purchased the ETX 80 BB and have recently purchased the DSI Pro with filters to begin imaging. The ETX 80 BB came with the Autostar #494. Can the 494 be used with the ETX 80 and DSI properly? All the information I find says the serial cable goes from the computer to the Autostar ( 494 does not have a port to plug into). Do I plug my serial cable into the AUX port on the telescope and to the computer? johnMike here: The DSI and LPI imagers work with the #497 AutoStar, not the #494. You could still use the DSI for imaging but without the AutoStar connectivity.
Mike, I have found instructions on how to use the 494. http://www.meade.com/manuals/TelescopeManuals/AutostarManuals/ 506CableSet.pdf Let me know if you see any problems here please. JohnMike here: With the #506 serial cable (which is different than the #505 cable that comes with the DSI) you CAN control the AutoStar #494 from a computer. However, the DSI software requires a #497 AutoStar.
Check the Feedback Archives for previous editions of the User Feedback pages.
Go to the ETX Home Page.