Last updated: 31 May 2007
Subject: dismanteling the ETX70 scope Sent: Saturday, May 26, 2007 04:28:19 From: email@example.com (firstname.lastname@example.org) start with going to "astro76.mypunbb.com/viewtopic.php?id=3958", which brings you to the Forum in general, choose " desmontando el ETX 70 por Piter", thats the series of about 50 fotos I followed to dismantle the scope. Once I get things OK with the decl.motor, i'll send you more details. Tell me if you got this right. andrew.
starting from foto n 21 from the link I send you , I unsrewed the 3 bolts holding the dec. arm with mecanism and electronics. After having undone the arm protection,( 5 screws), I loosened the printed circuit, controlled the dec and azimuth motors, which I found both OK so the problem must be in the circuit board itself, which I can't check because off my lack of knowledge in electronics. I don't have the drawing of the circuitery anyway. I cleaned the lot, I did not find any grease on the circuit board, and little on the mecanism. To asemble again, follow the explanations given in the link I send you. By the way, I fell in my shack, dropping the scope and tripod to the hard ground, ..I was lucky, no harm to lenses, only the plastic eyepieceholder got bend and I had to straighten it out again using a pair of players. If any information is unclear or missing I will come back on it OK ?. Thanks a lot. andrewMike here: I'm sure this will be helpful to others.
I have a "terestrian scope"for the moment, until I get to the problem seriously. The printed circuit on my scope says " ETX 60- C-2005 Meade- 15-6855-00 Rev 6. I don't think this is any worth to otrhers!!. Yes the fall was a lucky one, I got away with it with some black spots and some blood too. but not serouisly. Thanks for all. andrew.
Go back to the Tech Tips page.
Go back to the ETX Home Page.