Last updated: 21 June 2012
This page is for user comments and information specific to the Meade ETX-90RA (originally known as the "ETX Astro"). Comments on accessories and feedback items appropriate to other ETX, DSX, and DS models are posted on other pages. If you have any comments, suggestions, questions or answers to questions posed here, e-mail them to me for posting. Please use an appropriate Subject Line on your message per the Site Email Etiquette. Thanks. Remember, tips described on this site may invalidate the warranty on your telescope or accessories. Neither the submitter nor myself are responsible for any damage caused by using any contributed tips.
Subject: Baffle Sent: Sunday, June 17, 2012 10:37:40 From: Glen Perry (email@example.com) I have read the posts and cautions re: baffle reattaching. I have an ETX RA purchased around 1996. Meade will not work on the scope. I took the scope to my local Meade dealer. He reset the baffle in the original tape ring. (leaving a small scratch just outside the secondary). The baffle is slipping to the side again and I want to fix it before it drops onto the primary. Meade offered to replace the OTA for about $350 but I just can't afford that. Are there any suggestions you have before I attempt to replace it with epoxy as suggested on your website by Clay Sherrod? With what should I clean the inside of the miniscus? He said alcohol solution. What are the proportions? Thanks four your help. Glen Perry Sent from my iPadMike here: See the articles "ETX-90RA Secondary Baffle Fix" and "Re-attaching the Baffle" on the Helpful Information: Telescope Tech Tips page. The info there is all that I have.
Thanks. I did check those articles. Thanks for your prompt reply. Your website has been most helpful over the years.
Subject: broken off arm on ETX-90RA Sent: Saturday, June 2, 2012 13:25:14 From: Jon Iverson (firstname.lastname@example.org) I was hoping you might know or could point me out where on the site this might be located but I have a broken right arm on the ETX-90RA. It's broken right where it screws into the base. I glued it several years back and it held up, but it has come undone again and I'd like to repair it in a better way. Was hoping to catch the transit next week. Thanks in advance, JonMike here: You can check the Helpful Information: Telescope Tech Tips page but I don't recall an article that specifically covers reattaching an ETX-90 RA fork arm. Good luck.
Check the Feedback Archive for previous editions of the User Feedback pages.
Go to the ETX Home Page.